When a pH meter is reading incorrectly the simulator helps determine if the problem is related to the electrode or to the transmitter. It simulates pH and mV values: from 0-14 pH units in 1 pH unit increments (±414 mV). The simulator operates on a 9-volt battery and has a BNC connector, which attaches to the pH cable adapter.
The simulator isolates the transmitter by mimicking the load of a DO sensor. The 4-pin connector attaches directly to the DO cable, in place of the sensor. In doing so, the simulator can help confirm whether fluctuations in the transmitter display are caused by the sensor, or caused by problems with the instrument or cable.
Push button pH/mV selection
pH accuracy of ±0.05; mV range of ±2
USA/NIST pH buffer standards simulation
16 combination values
1 GΩ high input impedance test
Large display with bright LED
Protective rubber boot doubles up as stand
Power-saving meter runs on 4 'AAA' x 1.4 V batteries